Live Webinar How to Conduct a Human Factors/ Usability Validation
01:00 PM EDT | 10:00 AM PDT | 12:00 PM CDT Duration 60 Minute
Webinar Includes : All the training handouts , Certificate by the Speaker ,Q/A and 60 Min Live Webinar
"Hear By Edwin Waldbusser is a consultant retired from industry after 30 years in management of development of medical devices (5 patents). "
Human Factors/ Usability Validation
This webinar will explain the procedure described in ISO62366 and the 2016 FDA Guidance for a compliant human factors/ usability validation. HF/U validation is very different from device validation. For example, success criteria is qualitative rather than quantatative as is in device validation. Claiming success because eg. 95% of test participants did not commit a user error is not valid. Nor is 100% positive test results sufficient.
Why should you attend :
Following the implementation of the reults of a Human Factors/ Usability study, a validation of the safety and effectiveness of the use of the device must be conducted. We will explain the FDA required number of validation participants from each "distinct user population". We will explain how to choose the tests to be conducted and the studies that must be completed prior to the actual validation test. The post test participant inquiry is critical to validation success. we will describe how to do this.
• Required number of participants;
• Qualitative success criteria;
• Choice of tasks to validate;
• Post test participant inquiry
Who will Benefit
• Development Engineers
• Production Management
• QA/ QC personnel
• Software developers
• Engineering management
Industries who can attend
This 60-minute online course is intended for professionals in the Medical Device, Biotechnology,Pharmaceutical Industry. Although not presently stated in the draft , the same guide could be used by FDA Regulated Industries personnel.